The short answer is "yes," but we have a range of tips to help make sure this goes smoothly.
First, you will need an accurate reference for temperature and/or humidity. Each value is calibrated separately and you can choose to calibrate one, both or neither as necessary. Accuracy of the uncalibrated Meter can be found in the manual.
Calibration is only intended to correct for small variations (typically up to a few degrees or around 5% relative humidity). If you're seeing a bigger discrepancy than that while calibrating, please contact us at SwitchBot App Feedback (left side menu bar) or reach email@example.com for more assistance, as that would indicate that there's a problem with either the calibration process or your Meter. We'll be happy to help sort it out.
WITH A REFERENCE DEVICE
If you have a trusted thermometer or hygrometer that you know to be very accurate, you can use it as a reference. Simply place it next to the SwitchBot Meter in a stable environment, let them sit for at least 30 minutes to acclimate to each other, be sure you have an updated reading and then use the calibration controls (shown below) to adjust the SwitchBot Meter to match the reference.
To access the Meter settings, tap the gear icon in the upper-right corner of the Meter you want to control in the SwitchBot App homepage. Then click "Calibration".
If you don't have a reference device, you can still calibrate your Meter. Ice water makes a great temperature reference. Just mix up a bowl of ice water, with a good even mixture of ice and water. Give it a good stir and let it sit while you seal your SwitchBot Meter in a zip-top baggie. Get as much air out of the bag as possible. Place the SwitchBot Meter (inside the baggie) into the ice water mixture. It's best to leave the zip-top out of the water, just in case it doesn't seal entirely.
Give the SwitchBot Meter at least 30 minutes to acclimate, be sure you have an updated reading and then adjust the calibration controls so the value reads exactly 32.0°F (0.0°C)
If you have any difficulties, consider these tips:
First, make sure that when you run the test that you remove as much air from the bag as possible.
Second, make sure the side of the Meter with the writing on it is facing into the water.
The last thing is, to make sure the Meter is as submerged as much as possible because of the plastic case, the battery and the other components that need to be brought to equilibrium.
For calibrating humidity, the "salt test" is quite popular, and instructions can be found readily on the Internet. The chemical principle is sound and it can work quite well, but we find it to be difficult to execute consistently and accurately.
Calibrating to an inaccurate reference can be worse than not calibrating at all, so we prefer the Boveda One-Step Hygrometer Calibration Kit, available for less than $10 on Amazon.
Just open the baggie, seal your SwitchBot Meter inside and place it at room temperature in a location with minimal temperature variations. Let it sit undisturbed for at least 24-36 hours, be sure you have an updated reading and then adjust the calibration controls to read exactly 75.0% humidity. Keep the calibration kit sealed and you can use it multiple times for at least 3 months after the initial test.